Voluntary Field Correction
Mizuho, located in 2173-13 Ota, Sakura City Chiba, Japan, is engaging in a voluntary field correction for two of its microsurgical operating room tables. Tables affected are the 7300B and 7300BX. These tables are manufactured in Chiba, Japan and distributed by Mizuho America.
The voluntary field correction is due to two deficiencies in the two surgical tables. The first is tilt movement of the table after sitting idle for a long period of time and the second is unintended movement due to infrared operating signal errors while using the hand pendant in wireless mode.
Tilt Movement – Leaking Valve
- Tilt Movement – Leaking Valve, 7300BX (18-351-70) serial numbers, Quantity 7
MUC-20163-0001, MUC-2016Z-0002, MUC-2016Z-0003, MUC-20177-0005, MUC-20177-0007, MUC-20177-0008 & MUC-20177-0009
- Unintended movement due to infrared signal errors, 7300BX (18-351-70), Quantity 10
MUC-20163-0001, MUC-2016Z-0002, MUC-2016Z-0003, MUC-20177-0005, MUC-20177-0007, MUC-20177-0008, MUC-20177-0009, MUC-20189-0017, MUC-2018Y-0018 & MUC-2018Y-0019
- Unintended movement due to infrared signal errors, 7300B (18-350-70), Quantity 8
MUC-20168-0001, MUC-20168-0002, MUC-20171-0003, MUC-20171-0004, MUC-20173-0005, MUC-20173-0006, MUC-20178-0007 & MUC-20178-0008
Following is the recall plan for the Voluntary Recall reference